Das HP 8514B S-Parameter Test Set
misst alle 4 S-Parameter im
- Frequenzbereich von 45 MHz bis 18 GHz.
Es ist eine Ergänzung zum HP 8510 Network Analyzer.
ausgestattet mit Option H83:
Der Unterschied zur Standardversion:
20 W Peak Power
bis 18 GHz, oder "Normal Power" bis 20 GHz.
Bitte fragen Sie nach den zusätzlichen Kosten für eine rückführbare ISO Kalibrierung.Ein Datenblatt zu diesem Artikel finden Sie hier.Eine vergrößerte Abbildung befindet sich hier.
English translation of German description:
The HP 8514B S-Parameter Test Set
provides the capability to measure all four S-parameters of a two port device with a single connection over the 45 MHz to 18 GHz
The S-parameter test set architecture is ideal for measuring two-port devices where it is not convenient to physically reverse the device to measure the reverse parameters, or for networks that need to be adjusted while being measured with full error-correction employed. The test ports have rugged precision 7-mm connectors and may be adapted to other connector interfaces with the appropriate precision adapters. Along with an integrated, four-channel frequency converter, the test set includes two 90-dB step attenuators for changing the incident power level at the test port and two bias networks for applying DC bias to the test port center conductor in active device test applications.
The 8514B Option H83 is a standard 8514B test set that has been modified to allow
peak CW power levels of up to 20 watts
to be present at the test ports.
All other instrument specifications remain standard with two exceptions:
Exception 1 is in dynamic range specs: from 0.045 - 12.4 GHz the dynamic range is reduced by 2 dB and from 12.4 - 20 GHz the dynamic range is reduced by 4 dB;
Exception 2 is when the test set is configured for large signal measurements the operating range is limited to 0.5 GHz to 18 GHz.
The bias tees have been removed from this test set.
The added couplers (A21 and A22) are reversible for versatility in configuring special test systems.
Injection of a second signal for intermodulation testing of mixers, noise figure and spectrum analysis for amplifier testing are some examples.
Please ask for the additional fee for a current and traceable ISO calibration.Please find a corresponding data sheet here.Please find an enlarged image here.